AD frequency range: MHz. Square wave: MHz. AD adopts MHz Active Crystal. The module can output two sine waves and two square. IC AD is an ADI DDS chip which be common used in many DDS projects. This AD DDS module is basic on this ADI AD chip. This module provide. IC AD is an ADI DDS chip which be common used in many DDS projects. This AD DDS module is basic on this ADI AD chip.
|Published (Last):||11 February 2009|
|PDF File Size:||10.21 Mb|
|ePub File Size:||20.34 Mb|
|Price:||Free* [*Free Regsitration Required]|
Powered by the ubiquitous AD, so it will be easy to find code you modlue be able to adapt to your own project. Connect your favorite gear.
The result should be a sharp, straight line without any instabilities or wiggles. In practice, seconds is sufficient to show any instabilities. Be the first to review this product.
Only registered users can write reviews. The second modul speed ensures that nothing could possibly change state that could interfere with any measurement. Since it may not be easy for many kit constructors to carry out this adjustment, and there are already enough variables, it seems only polite for QRP Labs to supply the module ready-adjusted. Affordable SDR for everyone.
DDS module testing
For technical support, please open a ticket on Itead Support System. To test the frequency stability, a simple 30m band direct conversion receiver is used, powered from a well-regulated linear power supply. Where To Buy Nooelec. A great relief, actually There is therefore no squarewave output from the AD on-chip comparator.
The duty can be adjust by a adjustable resistor – which change the adjust voltage level of the comparer.
Join us on Moule. Features 2 channles square wave output and 2 chanles sine wave output Sine wave: Shipping Shipping Methods Delivery Information.
Turn the potentiometer clockwise “right” carefully, just a little at a time! Note that it is Ok to ignore ringing on the edges, this is not related to the potentiometer setting. Oscilloscopes Affordable portable and bench scopes–DIY kits available!
For more information regarding pin functionality please consult the datasheet linked to above. Forgot Password Login or registration. This is how the ‘scope trace looks when the potentiometer is turned all the way anti-clockwise to its limiting position.
Please, log in or ,odule. Be the first one to review this product. Note that for these tests, the U3 PA is actually disconnected.
Nooelec – AD 40MHz DDS Function Generator Module
This is fine, not a problem. Technical support For technical support, please open a ticket on Itead Support System.
Coming Soon QSX all-band all-mode transceiver. This is more expensive but the modules are evidently of a higher quality. Write Your Own Review Only registered users can write reviews.
AD9850 0-40MHz DDS Module
drs The image to the right shows 90 seconds of testing. AD Function Generator Module. What we are looking for is the absence of short-term instabilities that would show up as easily visible wiggles in this line.
The following examples are NOT correct oscilloscope waveforms: Effectively therefore, the kit is simply transmitting a continuous carrier at 10, The adjustment is almost correct here, but still slightly too clockwise.