BRAND offers the ideal pipette for every hand: BRAND’s new Transferpette® S. Transferpette® S models are are the perfect manual pipettes for demand-. Transferpette® electronic. F I R S T C L A S S · B R A N D. Gebrauchsanleitung. Operating Manual. Mode d’emploi. Instrucciones de manejo. 3. -8/ . Testing Instructions (SOP) testing of the piston operated pipette Transferpette®. BRAND additionally provides a calibration service (see chapter 7).
|Published (Last):||2 December 2016|
|PDF File Size:||19.30 Mb|
|ePub File Size:||6.95 Mb|
|Price:||Free* [*Free Regsitration Required]|
Trxnsferpette the pipette operation is clearly inaccurate, or if the instrument must be adjusted for solutions of different densities and viscosities or specially- shaped pipette tips, adjustments can be made using the Easy Calibration Technique. Parts will differ slightly depending on nominal volume of instrument. Comments to this Manuals Your Name.
Pipetting c Let the pipetting button slide back slowly.
Don’t show me this message again. It is beyond the scope of this manual to address all of the potential safety risks associated with its use in such applications.
Don’t have an account? For further information, refer to the accuracy table on pages 37 and Enter text from picture: Inspect the pipette tip cone for damage.
Beand the piston and seal for contamination. Safety Instructions Please read the following carefully! Limitations of Use Purpose The pipette is an air-displacement system for pipetting aqueous solutions with medium density and low to medium viscosity.
Limitations Of Use Limitations of Use Purpose The pipette is an air-displacement system for pipetting aqueous solutions with medium density and low to medium viscosity. Also See for Transferpette S Operating manual – pages. By pressing ‘print’ button you will print only current page.
Page 17 Inspect the pipette tip cone for damage. Table of Contents Add to my manuals Add. Hold the instrument vertically, with the sample in the tip for approximately 10 s.
Ensure that the tip is securely seated. The form can also be downloaded from www. Page 9 Pipetting Ejecting the tip Hold the pipette shaft over a suitable disposal container and press the tip ejection key to the stop.
Page 8 Pipetting c Let the pipetting button slide back slowly. ISO prescribes rinsing the pipette tip once with the sample liquid prior to the actual pipetting process.
Optimum analysis results can only be obtained with quality tips. Pipette tips transferpetfe disposables items! Don’t lay the instrument horizontal when the tip is filled. Warranty We shall not be liable for the consequences of improper handling, use, servicing, operating or unauthorized repairs of the instrument or the consequences of normal wear and tear especially of wearing parts such as pistons, seals, valves and the breakage of glass as well as the failure to follow the instructions of the operating manual.
Discharge sample a Place the pipette tip against the wall of the vessel. Limitations of Use The instrument is intended for the pipetting of liquids within the following limitations: Ask your supplier or manufac- turer for the form.
BRAND TRANSFERPETTE S OPERATING MANUAL Pdf Download.
Got it, continue to print. To do this, affix a tip, and aspirate a sample. To print the manual completely, please, download it.
Warranty Information Warranty We shall not be liable for the consequences of improper handling, use, servicing, operating or unauthorized repairs of the instrument or the consequences of normal wear and tear especially of wearing parts such as pistons, seals, valves and the breakage of glass as well as the failure to follow the instructions of the operating manual.
Autoclave the complete pipette without any further disassembling.
BRAND Transferpette S Operating Manual
Fitting the tip Use the correct tip according to the volume range or the color code. Checking the Gransferpette Depending on use, we recommend inspection of the instrument every 3 to 12 months. It is the responsibility of the user of this pipette to consult and establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
Set nominal volume Set volume to the maximum volume indicated on the instrument see page brandd for procedure.