The ELx continues our tradition of offering the tightest specifications of any microplate reader. Visit BioTek to learn more about our precision microplate. ELx Operator’s Manual. Revision History. Rev Date Changes. A. 1/96 First Release. B. 5/96 Revised reader specifications. Added Scanning. View and Download Biotek ELx operator’s manual online. Absorbance Microplate Reader. ELx Laboratory Equipment pdf manual download. Also for.
|Published (Last):||3 July 2012|
|PDF File Size:||12.75 Mb|
|ePub File Size:||13.86 Mb|
|Price:||Free* [*Free Regsitration Required]|
Re-download software basecode and assays. Optional Connect Printer 6: Reuse Of Standard Curves Select The Operating Environment As shown below, a small voltage selector card is located to the right of the fuse holder.
Biotek ELx808 Operator’s Manual
Page 8 System Test The kit instructions are provided so that users can see how it is possible to translate the kit wording into an ELx assay program. Autocal error, see error Introduction Chapter 1 Introduction This chapter introduces the ELx Absorbance Microplate Reader and describes its hardware and software features, and technical specifications.
Perform The Linearity Test test A Page If the above criteria are not met, the assay is invalid and the test should be repeated. The display updates to reflect the selection.
Microplate Readers: ELx Absorbance Microplate Reader – Overview
Selecting Elc808 Blank Map Page of Go. Perform Liquid Test 3 “uv” Models Only Reconfigure or replace fuses as needed see next page.
The starting ID is entered and automatically incremented by the software. The Startup Screen Clean Exposed Surfaces Errors and indicate that a motor was not able to move to its. This manual also for: Page Older-Style Module A failed fuse is usually an indication of another problem that a new fuse is not likely to remedy.
Run a system test. To avoid system instability, make sure the printer and reader are turned OFF before connecting them.
Add 95 mL of DI water; cap and shake well. Install the Filter Wheel The filters that ship with the ELx are installed in the six-position filter wheel.
Select the Operating Environment Page 34 – 3: Comments to this Manuals Your Name. The Louisville reference center recommends the following positive discrimination criteria according to the recommendation of the 2nd International Anticardiolipin Workshop. The ELx will perform a system self-test, displaying the screens shown below until initialization is complete.
Recommended Qualification Schedule If the replicate selection is 1, this option is not available.
BIOTEK ELX OPERATOR’S MANUAL Pdf Download.
Page 6 Attach the Cable Instrument Qualification Chapter 4 Instrument Qualification This chapter discusses the tasks and procedures necessary for qualifying instrument performance initially and on an ongoing basis. Assay Detail Report Dual wavelength readings can significantly reduce optical interference caused by scratched or fingerprinted microplates since the scratches or fingerprints reduce the amount of light on both wavelengths.
Page To define a cutoff formula for Positive and Negative calls: After the plate has been read, all requested reports will be generated.
Delta values are adjusted relative to the signal strength of the reference channel. Read Options 4-character alphanumeric name. Plate screen to select one wavelength.