FUELS & LUBRICANTS LAB. ENGLER’S VISCOMETER. 3. The water is taken into the water bath and the oil whose viscosity is to be determined is taken into. Asphalt engler viscometer is commonly used petroleum testing equipment. During the experiment, the sample should be a continuous linear flow. Engler Viscometer LEV-A10 · Graduated Flask: ± ml; Standard Water Value: 51 ± 1 s; Temperature Range: 0 ~ °C; Temperature Controlling.
|Published (Last):||19 July 2014|
|PDF File Size:||11.72 Mb|
|ePub File Size:||4.42 Mb|
|Price:||Free* [*Free Regsitration Required]|
The instrument should be cut off if the power failsask the professional and technical personnel overhaul and troubleshooting before they continue to use, to prevent accidents!
Enge La degrees K At the 2nd sample test has been engldrthe time is t2, then click the button, then the time is displayed in the display window around twice the average time: Emgler back to extractor http: The first two samples can be testedconsistent with the aforementioned method.
Contact Us apex lab equipment TEL: The handle cover operating table. LED electronic timing ; 10the heating power: Liquid by the external force in the resistance occurred between liquid molecules called viscosity.
Fifth, the engler viscometer,viscosity determination structural characteristics of the instrument 1Reset button: During the experimentthe sample should be a continuous linear flow. Sixth, the engler viscometer,viscosity determination use A test preparation 1, using the WNE-1B asphalt Enge La viscometer should be carefully read this manual.
Engler viscometer special corkbefore the testthe inner pot plugged orifice ; tests, pull the cork. Right channel test system data display window. Reading is the time t1, then chronograph buttonstiming window is displayed as Two using the method When used according engleg the following steps: Left channel test system data display window. Second, the engler viscometer,viscosity determination main technical characteristics 1, the instrument for the desktoponetwo inner pot structureinside and outside the pot all adopt stainless steel materialthe inner pot inside after special treatment, its smoothness meet test requirements.
Engler Viscometer (Digital Display) | Lab-Kits
Instrument chip computer control system reset buttonpress this buttonsingle-chip computer from the initial state to start the program. Inner pot mounted specimensthis cover to cover the inner pot.
Engler viscometer special measuring cupthe left and right channelsone each. Enge La degrees Asphalt specimens: Third, the engler viscometer,viscosity determination main technical parameters and indicators 1 Power supply: